sorry guys I'm pluggin away on them. I've got a friend who's comin over to help me assemble them since my time is so limited. Heck, I don't even have time to come on the forums much anymore lately. I'm sure you've all noticed my decreased presence on the forum.
yeah the ones with the range selection aren't done yet. Those require a little bit more time and effort so they are taking a little longer. I have had to redesign the enclosure (is slighly larger) so that the controls can all be on the front face.
for those that have purchased the meters with the range settings, I'm in the process of completely rethinking how to build these. I apologize for the longer wait. See, I didn't realize that by cutting the board down, I change the impedance of the output and therefore affecting the second range being displayed. I can get it calibrated to one laser but since it's not linear, a different powered laser is off quite significantly. So a 150mw green laser registers as 150mw, but when you point a 300mw red at it, it shows as a 500mw laser. I've got someone helpin me get this right though and should have it figured out soon.
All the other meters are done and will be shipping within a day or so.