Hi chloderic,Hi jnrpop, could you please post the startsequence of the device by using a serial-command tool , i would be interested in the selftest passing code at the end.
I have sent you such a tool, it is on the stick ....
Hello, yesterday I had to work and therefore could only look up today ...Hi chloderic,
I tried using the Te....ite software you placed on the usb, but I can’t seem to read the start sequence or anything when the spectrometer is connected. Is there a command line I need to enter into the Te....ite software to register the spectrometer and the correct port?
O, I also got the Spectrometer to register again on Moscs software and the original B&W software, but still get the same message when using Speactral Studios.
It could be possible the Spectral Studios software we have has a license time frame or error with registration?
Dear Agastar,Thanks, the printer is a clone of an ultimaker 2+ extended. I had an original UM2 a few years ago and fell in love with them. These days it is hard for me to justify the cost of the true UM2s I think the layer height was .15 for this print (cura default).
The alignment of the fiber optic connector did not cause me any issues. The power connector on my spectrometer lined up with the left side of the hole, however the connector housing on the cable was bigger than the hole. Even if it had lined up in the center, this cable wasn't going to fit so I had to enlarge the hole post-print. I've seen barrel plugs with much thinner bodies so for me, this wasn't a design issue
Hope that helps